Customer Service: 1-800-331-7993
August 29, 2014
There are 2 basic types of attachments used with slow-speed motors: nosecones and contra angles.
A nosecone is a straight attachment that will accept a slow-speed bur or a shaft-driven angle (contra angle or prophy angle). Nosecones are unique and come with different gear ratios. The default is a 1:1 ratio – the nosecone will operate at the same speed as the motor. 4:1 is a common gear reduction, the nosecone will spin at ¼ the speed of the motor. Some nosecones (primarily those designed for use with electric motors) will also have speed increasing gears, so they will operate at a 1:5 gear ratio (for example), or 5 times the output speed of the motor.
Nosecones are standardized so they all will accept the same diameter bur or shaft driven attachment. Nosecones also incorporate a pin of some sort to prevent rotation of any shaft driven attachment placed on the nosecone. All shaft driven attachments have a groove that slides over this pin.
As nosecones can accept a slow-speed bur, all you may need to perform a particular procedure may be a motor and nosecone (and bur, of course).
The other type of attachment, a contra angle, will work with gear driven attachments only (most commonly some sort of head). They will not accept a bur, so further attachments are required. As the name implies, a contra angle provides an angle for the next attachment which can improve intra-oral access.
Both Midwest and E-type contra angles accept the same type of gear driven heads. The heads incorporate a drive shaft with a gear at the end that seats into the contra angle meshing with the internal drive shaft causing the head to spin. The drive gear has pointed teeth making it easier to seat the two halves together. The head also has square “teeth” under a threaded collar that mesh with the square “teeth” on the outside of the contra angle. These teeth hold the head onto the contra angle and prevent the entire head from spinning (so only the drive shaft spins). It is these teeth that one must count to determine compatibility between a head and contra angle. Heads and contra angles come with either 12 or 14 locking teeth.
Star systems do not normally use a contra angle attachment. Instead, they use a straight attachment which accepts a Star-specific head. Star heads have an elbow incorporated at the end to provide the angle normally provided by a contra angle attachment as used by other systems.
American Dental Accessories, Inc. also has an after-market contra angle that will work with a Star-type motor. This angle will allow you to use standard heads with your Star system (which can save money over the more costly Star-specific heads).
Regardless of system, a contra angle (or angle attachment) will require a 3rd attachment for use with a rotary instrument and will not be a complete set-up for a given procedure (as a motor and nosecone alone can be).
Finally, there are heads. As mentioned above, heads will have both drive teeth and attachment teeth (or drive teeth and a threaded elbow). The number of attachment teeth will determine compatibility with a particular contra angle. The head will accept the rotary instrument with which one will perform a given procedure. The most common head is a latch head which will accept a latch (or RA, for “Right Angle”) bur. RA burs have a groove at the end into which the latch of the head will secure holding the bur in. Some heads also accept standard friction grip burs, exactly as used in a high-speed handpiece.
Other heads are designed only to accept prophy cups. Prophy cups can come with either a threaded “screw on” shaft or that simply “snap on” a knob designed for this purpose. Some are also attached to a standard latch-type shaft so they’ll work in a standard latch head.
The flexibility afforded by the various head configurations allows for a tremendous range of applications for a slow-speed set-up. This flexibility can allow for great value with a slow-speed system.
June 29, 2014
The turbine is the heart of the high-speed handpiece and bearings are the heart of the turbine.
Bearings are the most common failure point of a turbine and are often the primary differentiator between one turbine and another.
Way back in Tech Tips #22, “High-speed Handpiece Design,” we covered all of the components that make up a complete bearing assembly. For ease of reference, we’ve included the diagram of a bearing assembly below. For further explanation of the components, check out Tech Tips #22.
Today, we’re going to look at some of the different bearing materials and designs on the market.
Many turbines use stainless steel bearings that require lubrication. The balls, inner ring, outer ring, and shield are all made of stainless steel. The ball cage will be made of a polymer (there are a few more variations within this broad category and different types of polymers for the cage, but all these permutations require lubrication).
Stainless steel bearings have been in use for decades and are a proven design with good performance and good reliability. They are manufactured in large quantities for dental turbines and many other industries so cost is comparatively low. As these turbines incorporate metal bearings in metal housings, they require lubrication. Most handpiece lubricants on the market are designed to withstand the rigors of sterilization, but these turbines should still be lubricated every time they are used (see our handpiece maintenance products here). Lubrication after using and before sterilization is generally adequate, but consult the manufacturer of your turbine and lubricant to determine if post sterilization lubrication is required as well.
Many turbines currently on the market are advertised as “lube free.” There are 2 primary methods of manufacturing lube free bearings:
LTL bearings are still the same basic stainless steel design so they share many of the features of standard stainless bearings. The greasing and sealing process adds to the cost (and they aren’t manufactured in quantities like the standard bearings), so they will add to the cost of the turbine or handpiece that uses them. The sealing process, also, prevents debris from getting into the center of the bearing assembly and on the actual steel balls, so maintenance is a little easier. Nonetheless, these bearings (or, more accurately, turbines that incorporate these bearings) still need to be cleaned after every use and before sterilization.
Last of all, there are ceramic bearings. Ceramic bearings are actually made of a ceramic silicon nitride, so they have a very smooth low-friction surface. The low-friction surface removes the need for lubrication and also minimized heat build-up during use. They, also, withstand high temperatures very well, so they withstand repeated sterilization better than stainless steel bearings.
Both LTL and ceramic bearings will run at higher rpm than stainless bearings and are better able to handle higher air pressure (they typically require in excess of 40 psi drive air pressure). The higher rpm can help these turbines cut faster so some practitioners feel they perform better (naturally, this is subjective).
In summary, the materials and techniques used to manufacture lube free bearings are more costly than standard stainless bearings, so these bearings (or the turbines that incorporate them) often cost twice as much as turbines that need to be lubricated. It’s up to the practitioner if the advantages are worth the expense.
April 30, 2014
Before we get started, we want to reiterate our number one objective with Tech Tips (and when you call us): to empower your practice with basic maintenance, installation, and repair knowledge, so that you're not paying full-service distributors exorbitant fees on a regular basis. While taking the time to familiarize yourself with your equipment will have a learning curve, you'll be glad you did. Once you get it down, you'll be saving lots of money and will feel more in control of your practice. Without further adieu:
Order today and receive a $150 Merchandise Credit! Call 800-331-7993 to place an order and mention the code: TECHTIPS (order must be placed by May 5th, 2014)