Customer Service: 1-800-331-7993
September 29, 2010
X-rays are extremely important diagnostic tools integral to treatment planning. Practitioners commonly have trouble with films coming out of the processor too light, too dark, or with an otherwise poor appearance.
This months issue of Practice Tips discusses many of the common problems with x-ray films and the solutions to resolve them.
Step 2: Check the settings on the control. Most of the later model machines have only a timer setting. Turn the timer up several steps and try again. If you have an older model with mA, KVP, and timer controls, make sure that these are where they ought to be. Higher KVP settings will result in darker radiographs, mA affects density. Do not just start turning knobs. Always go up with just one control at a time.
Step 3: Make sure that you are holding the exposure button down until it cuts off. X-ray machines are required to have “dead man’s switches” so that the emission of radiation stops when the button is released. Occasionally an operator will be in a hurry and release the button early. If this happens, you will get a light film.
Step 4: Several things in the darkroom will cause a light film. Cold solutions in the processor or dip tank, weak or depleted solutions, and film being processed too rapidly. Ideal temperature for solutions is 80°-82° F. The solution should be changed at least monthly, more often for busy practices. A chart displaying date of change is a must. Even better, put a calendar next to the processor to record all maintenance complete with check boxes to verify it has been performed.
Most automatic film processors have some sort of solution heater (usually a pad that goes under the tanks or some sort of bar/probe that goes into them). If the solutions are cool, the solution heater is usually the cause. You will need a multi-meter to check for power or appropriate resistance of the heater. Specifications will vary depending on the make and model of processor. (Some developers also have a separate temperature sensor – see below under “films too dark” for more info on these). Consult your owner’s manual or call American Dental Accessories, Inc. and our tech support staff can assist you with a determination.
Step 5: Check the speed. 4.5 minutes is prescribed. If running too short– the speed can often be calibrated. Consult your owner’s manual or call American Dental Accessories, Inc. tech support line and we can assist you with calibration. Worn or dirty transports may not move freely and could slow down processing as well.
Step 6: Contaminated chemistry. Many times contaminated chemistry causes poor quality film. Thoroughly clean all tanks and racks/rollers and change the chemistry. Rinse everything with distilled water.
Evidence of contaminated chemistry:
• Dark developer solution.
• Black deposits in the bottom of the developer solution pan/tub.
If these problems exist the processor must be cleaned thoroughly, rinsed with distilled water, and the chemicals replenished.
Note: Depleted chemistry can also cause light films. Replenish your chemistry with 6–8 oz. solutions daily. During the day you may top off the solutions with distilled water (1-2 oz. only– more will dilute the chemistry).
The Air Techniques AT2000 processors have the temperature sensor built in to the solution heater so you will need to replace the heater if the sensor is bad.
Note: “A friend in need….” If you still cannot get to the root of the problem, and help is not available, try this: Using your x-ray and your film, take a picture. Go to a nearby dentist’s office, (if pan film, you must be extra careful to protect it from light). Process the film with his/her processor. If you can, take a film at his/her office and bring it back, and process it in your processor. This should really tell you where your problem lies.
NOTE: The machine that is taking the x-rays does not cause dark films. Dark films are only caused by the chemistry being too warm or light leaks. The slightest crack in the safelight filter will cause dark pan or ceph films (disregard periapical films, pan & ceph films are 200 times more sensitive to light). A very slight leak under or around the door (or into a daylight loader) will cause dark films. To test for light leaks, lay a fresh film out on the counter, place a coin on the film, wait a minute or two and then develop the film. If you can see the image of the coin, you have light leaks. If using a daylight loader, try using the processor in the dark. If this corrects the problem, your daylight loader leaks. To check for light leaks around a darkroom door, stand in the middle of the darkroom and turn all the lights off. Close your eyes for 60 seconds. Open your eyes and look for the source of light.
Also, make certain you have a safelight with a GBX filter. Many times one may be tempted to use a red bulb but this is not a true safelight and could expose your films. Using a dedicated safelight with a low wattage bulb and separate GBX filter is essential to film quality.
Last of all is the film itself. Make certain that your film is safely stored and is not expired. Always make certain to rotate your stock so that the oldest film is used first. When re-stocking film (and supplies in general), always put the newest supply behind the older stock on your shelf.
There are additional checks and possibilities for panoramic films as well. These are detailed in our equipment repair and maintenance guide Doctor, Did You Check the Breaker Too? available from American Dental Accessories, Inc. in hard copy or on interactive CD.
American Dental Accessories, Inc.
We offer every day low prices, free technical support, and friendly service 64 hours a week. Call 1-800-331-7993 or order online now!
©2010 American Dental Accessories, 7310 Oxford Street, Minneapolis, MN 55426.