Customer Service: 1-800-331-7993
May 25, 2010
You use your handpiece every day. It’s one of the most important pieces of equipment in the dental office, but how does it work?
As you probably know, a typical highspeed handpiece uses air to rotate a cutting bur at about 350,000 rpm’s. These instruments are among the fastest spinning turbines in the world.
Midwest, Star, KaVo, NSK and Lares own about 90% of the US handpiece market although there are over 50 different makes and models available from various manufacturers throughout the World. Nonetheless, all of these manufacturers use the same basic design.
The typical handpiece is connected to an air hose. As you press on the foot control, air enters the handpiece through the back. The air moves up the body of the handpiece within a small tube. The air then enters the head of the handpiece where the turbine is housed. It is the turbine that performs all the work of the handpiece. The handpiece itself is just a handle to provide a means of controlling the turbine (as well as serving as a conduit for air to drive the turbine and air and water to cool the surface being cut). In fact, in manufacturer circles, a handpiece without a turbine is called a “shell”- as it’s really just a hollow case. The heart and soul of the handpiece is the turbine. The standard components of a turbine are shown in the following diagram:
At the center of the turbine is the spindle (#1 in the diagram above), this is the shaft that spins. Inside the spindle is the chuck (#2 in the diagram above), this is a hollow tube that holds the bur, which of course does the cutting. As you can see in the diagram, the chuck has little slits at the end. These slits allow the chuck to compress as it moves forward inside the spindle to grip the bur. On standard chuck handpieces (like that shown above) the chuck is threaded (as is the inside of the spindle) and a bur wrench is used to rotate the chuck screwing it farther into the spindle and compressing the end. For autochuck handpieces (e.g. push button) the chuck is not threaded, but rather spring loaded and includes a mechanism that allows manipulation of the chuck forward or back to open and close the end to grip (or release) the bur.
Placed on the middle (from front to back) of the spindle (it’s actually pressed in place) is the impeller (#4) this is what catches the air causing the turbine to spin.
As you can see the spindle and impeller are fairly simple solid pieces of metal (usually stainless steel or aluminum) and thus, rarely fail. Chucks will occasionally fail as they can fracture where the slits are, but this is still relatively uncommon.
On either side of the impeller are bearings (#’s 3 & 5 above). But aren’t those solid cylinders in the diagram? Bearings are little metal balls, aren’t they? Well, yes and no. Technically, what we have are bearing assemblies; these are what allow the turbine to spin (by reducing friction). The following diagram is a cross-section of a typical bearing assembly:
As you can see, the bearing assembly, or “bearing” as referred to in the handpiece world, consists of several components (including the balls you expect). At the heart of the assembly are the actual ball bearings. These are generally stainless steel, although ceramic bearings are being widely used as well. The bearings ride around an inner ring (as it’s labeled in the diagram) called the inner race. There is a slight groove (raceway) on the outside of this ring that the ball bearings roll in. The inner race is generally the strongest part of the bearing assembly as it is what is pressed onto the spindle.
There is then a cage that holds the balls at equidistant intervals around the inner race. This keeps the turbine balanced. Think of an old washing machine with the clothes all on one side. The “clunk clunk” of the washer is caused by the erratic rotation due to improper weight distribution. The cage keeps the turbine spinning smoothly as it maintains an evenly distributed load (more on load later).
Bearing cages can be made of a variety of materials, generally, they are a polymer (i.e. plastic) and are usually the most fragile component of the entire turbine. The plastics used are very resistant to heat and have very low-friction as they directly contact the actual bearings.
Outside of the cage is the outer race. Just as with the inner race, there is a groove in which the bearings ride. The outer race rests on o’rings (the two #8’s in the turbine diagram) which in turn rest against the inside of the handpiece head.
The entire bearing assembly is then closed off with a shield (as shown in the diagram). The shield is designed to keep debris from entering the cage and getting onto the bearings. Naturally, if debris gets onto the bearings it can throw the entire turbine out of balance, putting stress on the cage which then fails allowing the bearings to “clump” and causing the turbine to spin erratically (and eventually stop spinning completely).
All of these components are extremely small and thin. Whenever replacing a turbine, make certain you have removed all of these components of you old turbine from the handpiece. It’s very common for an o’ring, outer race, or shield to adhere to the inside of the handpiece head (or end cap). Note also, that all of these components are simply pressed together using a bearing press and held together with friction. This is why a press is required if you wish to repair your own handpieces, saving you money.
There are two more components to the turbine above, first of these is #9 a wave (thrust) washer– sometimes called a loading spring. The Quiet Air turbine diagram shown above is one of the only standard chuck turbines to use a washer. Washers are normally only used on autochuck handpieces to provide lateral tension, keeping the turbine pressed against the end cap to facilitate actuating the chuck. If you have trouble after installing a turbine that uses a washer, sometimes you can remove the washer and have the turbine function well. It is not uncommon (particularly with age) to get a slight build up within the handpiece head or to have the push button or other mechanism that actuates the chuck start to vary some from the original specs so the washer can impede proper function. The washer above is bent to provide additional tension, sometimes the washers used are flat and simply serve as spacers.
#7 in the diagram is the end cap. Technically, this is not part of the turbine. The end cap simply threads onto the back of the handpiece head to hold the turbine in. The end cap will also include the button (or lever) that is used to activate the chuck on autochuck turbines. The back bearing will usually seat into the end cap of the turbine. To simplify installation, some manufacturers include end caps with their turbines, providing them already seated on the rear bearing of the turbine. Whether or not an end cap is included with a turbine will not have any effect on the function of the turbine itself. It’s a matter of convenience and it can impact the price, of course. As mentioned previously it is extremely common for components of worn turbines or bearings to get stuck in the end cap, so always inspect the end cap thoroughly when replacing a turbine. Also, many end caps (like the Quiet Air shown) have a groove into which one of the o’rings will seat. Always replace this o’ring when installing a new turbine. You will probably need to use a scaler of explorer to remove the old o’ring.
As you can see, turbines are fairly complex assemblies, particularly the bearings.
Now that we know how a turbine is made – what does this mean to you?
With the insight we have into design, now we can look at what causes failure.
As already discussed, generally it is the bearings (specifically the bearing cage) that will fail first in a typical turbine. What causes bearing failures are as follows (in order of frequency/likelihood):
“Load” as the name implies, is weight or pressure. “Side load” is pressure applied to the inside of the bearing assembly in a perpendicular direction. Side load is usually the result of cutting with the side of the bur. “But I cut with the side of the bur all the time- can’t you do a crown prep?” Now we get into a more subjective area.
Naturally, cutting with the side of a bur is an everyday occurrence and should not damage your turbine or bearings. The key is that you allow the bur to cut. If you find yourself leaning into the bur, putting a lot of pressure on the side of the bur (say 5-10 pounds or so) this is creating side load. If you press on the end of the bur, the opposite end (which is in your turbine) will be pressed in the opposite direction (like a lever). This pushes on the inner race of the bearing compressing one side of the bearing assembly creating localized increased friction, which leads to imbalanced rotation which stresses the cage ultimately causing fracture and failure. Remember your Newtonian physics: for every action there is an equal and opposite reaction- the more you press on your bur, the more you press on your bearings.
Ideally, you need to allow the bur to cut and simply follow the bur as it does so. Do not press on the bur at all. The reality is that some pressure is required, but try to keep it fairly light to avoid stressing your bearings. Of course, worn or dull burs won’t cut as well, so they are more prone to induce side load. When side-cutting, always try to use a sharp and fresh bur or diamond.
The good news is that excessive side load will not necessarily lead to immediate bearing failure. Occasional pressure is expected and should not have a dramatic effect on turbine life. However, constant or frequent pressure certainly will. For some, saving a minute on a crown prep is worth the cost of more frequent turbine replacements, this is something you need to decide for yourself.
Generally, side load will have a greater impact on the rear bearing than the front bearing. If you replace your own turbines and notice that it’s usually the rear bearing that falls out in pieces (while the front bearing appears ok), you might consider trying a lighter touch when cutting with the side of your bur.
Using a bur that is bent or that is not fully seated into the turbine can also cause excessive side load. Naturally, a bent bur will not maintain concentricity and will wobble, stressing the bearings.
Using a bur that isn’t fully seated can cause side load as the weight at the front of the turbine is less than that at the back. This will not have as much of an impact on balance, but can still stress the front bearing (as it is normally doing most of the “work”). On standard chuck handpieces, this can also cause the chuck to fracture.
Excessive temperature should be self-explanatory. All dental handpieces are now made to be heat sterilized in accordance with the CDC. They should be sterilized in a steam or chemical vapor sterilizer at a maximum of 275° F (135° C). Dry heat sterilizers operate at much higher temperatures and should never be used on handpieces. You should check your sterilizer regularly for overheating with a lag thermometer. Generally, if your sterilizer is overheating, you will probably have damage to other items in the office, but it is not unusual to notice it in your handpieces first.
Most manufacturers recommend running their handpieces at about 35 psi (for highspeeds). Generally, anything in excess of 40 psi will cause damage to the bearings. Always check with the manufacturer for their recommendations, of course and try to stay within 5 psi. Some practitioners like to use air pressure over 40 PSI to achieve better cutting power. Remember, these bearings are very, very small and their components (especially the cage) are fragile. While the cage may not fail right away, you will experience premature failures, shortening the excepted life of the cartridge if you use higher air pressure than recommended. As with side loads, everyone must decide the value of increased speed vs. shortened turbine life for their method of practicing. While higher pressure will normally result in higher rpm’s, it will also usually result in shorter bearing (turbine) life.
Last of all, debris buildup inside of the bearings can cause premature failures. Debris will accumulate on the bearings causing the turbine to become imbalanced, stressing the cage and other bearing components leading to failure. The shield helps protect the bearings but debris can still accumulate on the outside of the turbine and imbalance it as well. Regular and proper maintenance is crucial to handpiece performance and longevity. See our issue of Practice Tips devoted to maintenance for more information.
Turbines are assembled under very tight tolerances all of which are to assure concentricity is maintained. All of the causes of failure will negatively impact concentricity in one way or another. Once again, the bearing components are very small and actually fairly delicate, particularly when you consider the speed of 350,000 rpm (or more). It doesn’t take much stress before they can be damaged.
Never run cold water over handpieces to achieve rapid cool downs after sterilization. Rapid cool downs could cause warping of handpiece components and you could also introduce contaminants onto your sterilized instruments.
You should never run your handpieces without burs. Doing so could cause damage to the spindle/chuck assemblies. This is more important with standard chuck turbines.
For autochuck handpieces (push button or lever) lubricate the chuck at least once a week to keep it clean and functioning well. Debris can clog the chuck and interfere with holding the bur.
A Quick Note on Electric Handpieces:
Most electric handpieces use similar designs to slow speed attachments. That is, there is a drive shaft which meshes directly with a gear in the turbine assembly, rather than an impeller that catches air. They still have bearings and, as discussed above, bearings are what typically fail. The same precautions apply to electric handpieces as pneumatic handpieces when operating (other than air pressure, of course). Beware of side loads, keep them clean, don’t use bent burs etc.
With electric handpieces, it’s even more important to avoid damaging your bearings as the electric motor will continue to yield the same rpm’s and torque at the bur even if your bearings have failed. To maintain this performance, however, an electric handpiece will need to work harder and can overheat. See the FDA's article on "Patient Burns from Electric Dental Handpieces" for more detail. Be sure to handle your electric handpiece with care too.
American Dental Accessories, Inc.
We offer every day low prices, free technical support, and friendly service 64 hours a week. Call 1-800-331-7993 or order online now!
©2010 American Dental Accessories, 7310 Oxford Street, Minneapolis, MN 55426.
May 19, 2010
In 2009 alone, Americans spent $10.5 billion on cosmetic solutions, so it’s no wonder why dental offices have moved to tap into this market. Nonetheless, There is a blurry line between offering cosmetic solutions and routine check-ups. The Washington Post reports that many patients are not responding well to smile makeover pressure during visits, and for good reason. With only about 40% of the US population having dental insurance and 15% under/unemployment, many patients are reluctant to spend $5,000 on treatments like Invisilign. Recall the feeling that you may get when dental reps employ aggressive tactics to sell CEREC machines or other dental equipment that may or may not improve your overall practice. Pressure can alienate, and while many dentists seek dental equipment from less pushy dental sales reps, the same can happen with your patients. So what can be done?
Given the subjective nature of cosmetic treatments, it's best just to educate and leave the decision up to the patient. Make sure every employee in the office is aware of your no-pressure policy, since in a lot of cases the alienation may not come from the dentist. If the practice is under pressure to increase the bottom-line, the best way to accomplish this is by the following:
1) Market the practice - the most inexpensive and effective way to advertise is online through Search Engine Optimization (focusing on keywords like, "Dentist in insert-your-city-here" )or third-party patient/customer resources (e.g, encourage satisfied patients to write reviews on sites like Yelp).
2) Empower yourself - Instead of being at the mercy of large dental supply companies, learn simple maintenance and repairs that can extend the lives of your dental equipment, like handpieces, chairs and units. On-site dental support can cost $100-$150 per hour, so cut down on unnecessary service calls. Remember American Dental Accessories offers complimentary technical support at 800-331-7993.
3) Rethink Dental Supply Purchases - While it's important to have a relationship with a dental supply company that you trust, that doesn't mean you need a rep that frequently comes into your office. Shop around and try the best dental supply company that works for your practice (you may find that several fit the way that you operate). For example, American Dental Accessories has recently started selling bulk toothbrushes that rival leading brands at over 40% below other dental supply companies--this alone can save a practice $1,000's a year. Moreover, stay dynamic with dental equipment and supply purchases--after all, there might be a better option out there for your practice.
May 12, 2010
The BBC reports that dentists might be able to play an integral role in helping to control blood sugar levels by treating gum disease. Edinburgh University scientists believe that the bacteria in the mouth, which increases inflammation, may inhibit insulin effectiveness, thereby increasing blood sugar levels.
Since over 24mm people (and growing) in the US have diabetes, this is a good fact to know since gum disease is a relatively easy condition to treat. Further evidence to support the body's overall health in relation to oral hygiene.