Customer Service: 1-800-331-7993
April 21, 2010
In a previous issue of Tech Tips, we de-coded the Statim error codes. Additionally, you may notice problems that won’t always trigger an error code.
The following are errors that can be observed which will not necessarily yield a code:
As we discussed, a number of error codes (as well as non-coded problems you can observe) indicate problems in the Solenoid Valve and the Pump. We’ve included some additional information for the Pump and Solenoid below:On a regular basis, check the bubble level – the bubble in the level on the Statim should be at the 5 o’clock position.
NOTE: When removing the Solenoid Valve stem, it is suggested by Sci Can to use a pair of pliers on the welded area of the valve stem (plunger tube). If done incorrectly, the pliers could collapse the hollow area of the valve stem, thus preventing the plunger from moving freely. However, we offer a spanner nut which provides an easier and safer way to remove and install the valve stem. By using the spanner nut with a wrench, the valve can be removed without disturbing the plunger.
STEP #1: Disconnect the power to the sterilizer by unplugging the power cord – as with all electrical devices, never rely on just the power switch. Disassembly of the solenoid valve with power on will cause valve coil damage, so it is imperative that you disconnect the power. Once disconnected from power, remove the nut, yoke, and coil.
STEP #2: Place the Spanner Nut over the valve stem (plunger tube) and align the Spanner Nut pins with the base of the valve stem (see figure 1).
STEP #3: Using a wrench on the Spanner Nut, loosen the valve stem to access the internal parts of the Solenoid Valve.
STEP #4: Remove any debris found in the valve body and blow air through all the orifices. Clean the plunger and ensure it slides smoothly in the plunger tube. Reassemble the solenoid valve.
You must use a special pump tester bottle, which is available through American Dental Accessories.
NOTE: The American Dental Accessories, Inc. tester bottle has a “swivel fitting” on the lid that allows you to control the position and balance of the bottle. In addition, we include 8” of tubing already connected to the swivel fitting which allows you to move the bottle a safer distance away from the machine during testing.
STEP #1: Disconnect the pump outlet flow gauge tubing (see Figure 2) from the boiler by using a 3/8” wrench on the compression nut while supporting the boiler fitting with a 7/16” wrench. Connect the Flow Gauge Tubing to the Pump Tester Bottle male connector and tighten.
STEP #2- Purge Cycle: Activate the pump for approximately 2-3 seconds to purge any trapped air from the tubing and to allow an even flow of water into the bottle. Empty the bottle of any water that may have entered it during the purge cycle. The bottle must be empty before proceeding with the test.
STEP #3- Test Cycle: Before activating the pump for the Test Cycle, have a stop watch or another timing device available for use during the cycle. Activate the pump and note the amount of time that it takes for the water level in the bottle to reach the minimum fill line. If the water level reaches the minimum fill line within 22-28 seconds, the pump does NOT need to be replaced and skip Steps #4 and #5. If the water level does not reach the minimum fill line within the specified time, check the inlet and outlet filters and proceed to the next step.
STEP #4- inlet & outlet filter cleaning (see Figure 3): Using hemostats or a tubing clamp, clamp off pump inlet tubing from the reservoir. (Note: It is important to crimp the tubing so that the reservoir does not drain.) Remove the flow gauge tubing from the pump outlet side using a 7/16” wrench. Remove the filtered elbow fitting and filtered barbed fitting from the pump. (Caution: Use a wrench on the pump body while removing filtered fittings.)
Clean the filter mesh of both the inlet and the outlet fittings by placing the fittings under running water, then blowing both ends out with an air hose. Reinstall filtered fittings using Teflon® tape.
Repeat Step #2- Purge Cycle and Step #3- Test Cycle.
STEP #5: If the water level does not reach the minimum fill line within the specified time, replace the pump.
The pump tester bottle and solenoid instructions above were taken from our Statim repair kit. This kit includes the pump tester bottle and the spanner nut as well as a number of other specialty tools and further instructions on Statim repair.
American Dental Accessories, Inc.
We offer every day low prices, free technical support, and friendly service 64 hours a week. Call 1-800-331-7993 or order online now!
©2010 American Dental Accessories, 7310 Oxford Street, Minneapolis, MN 55426.